Vascular Intervention // Peripheral
Balloon-Expandable Cobalt Chromium Stent System/0.014”/Rx
Proximal gold marker for superior visibility to support accurate stent placement
The gold plated proximal stent ring element facilitates superior visibility allowing accurate ostial stent placement.
Cobalt chromium alloy combining a lower profile with improved radial force
The cobalt chromium alloy thin strut stent design permits a low crossing profile, while maintaining radial force for vessel scaffolding.
Double helix stent design for high flexibility
The double helix stent design ensures high flexibility, homogeneous wall coverage and superb conformability.
proBIO coating for improved stent surface biocompatibility1
The proBIO silicon carbide coating acts as a barrier between the metal stent and the surrounding tissue and blood, protecting the surface of the stent. By providing a barrier against ion release, the coating creates a surface that reduces platelet aggregation while facilitating endothelialization.1
Indicated for improving arterial luminal diameter in patients with clinical symptoms attributable to atherosclerotic stenosis of the renal arteries.
|Dynamic Renal Stent|
||Cobalt chromium (L605)
||120 μm (ø 4.5 - 5.0 mm)
140 μm (ø 6.0 - 7.0 mm)
||proBIO (Amorphous Silicon Carbide)|
||Proximal gold marker
||ø 4.5 - 7.0 mm; L: 12 - 19 mm
|Catheter type||Rapid exchange (Rx)|
|Recommended guide wire||0.014"|
|Tip||Soft, short and tapered|
|Balloon markers||2 swaged markers|
|Shaft (proximal)||Hydrophobic coating|
|Nominal pressure (NP)||10 atm|
|Rated burst pressure (RBP)||15 atm (ø 4.5 - 6.0 mm)
13 atm (ø 7.0 mm)
|Balloon Diameter x Length (mm)|
|Rated Burst Pressure||atm3||15||15||15||13|
1 Rzany A, Schaldach M. Smart Material Silicon Carbide: Reduced Activation of Cells and Proteins on a-SiC:H-coated Stainless Steel. Progress in Biomedical Research 2001; May: 182-194.